*Update* Advisory has been corrected with the release of Dexterit-E 3.9.1
———————————————————————————————————————————————————————————————————————Advisory posted 18-Jun-21, re bug in the NI-6229 block
- The bug affects those sending out digital out pulses only AND running MATLAB R2019b
- Analog output is not affected.
- This bug will be fixed with 3.9.1 expected early July 2021.
- The workaround is to unlink the Sync TTL block, copy the BKIN 6229 DO block inside it and use that instead.
- Please contact email@example.com if you need help with the workaround.